| || |
|The transducer comprises a sensor element (shown in blue), a signal converter (at the right of the diagram) and a position magnet. |
The signal converter applies an electrical pulse to the sensor element and simultaneously starts a timer.
The magnetic field generated around the sensing element by the electrical pulse interacts with the field of the position magnet and produces a mechanical pulse.
The mechanical pulse travels back down the sensing element where a sensor in the signal converter detects it and stops the timer.
As the speed of the mechanical pulse is constant, the time taken for the pulse to reach the sensor can be used to accurately measure the position of the magnet.
Typically the mechanical pulse travels at 2800m/s and the cycle process is repeated 1000 times per second.
|All dimensions and specifications are nominal.
|Due to our policy of on-going development, specifications may change without notice. Any modification may affect some or all of the specifications for our equipment.
|USA & Canada
2216 Pottstown Pike
Pottstown, PA 19465
|Rest of the world
RDP Electronics Ltd
Grove Street, Heath Town
Wolverhampton, West Midlands, WV10 0PY
Tel: +44 1902 457512
Fax: +44 1902 452000